Can the LUTRON CM-6010SD log both Current and Voltage Simultaneously to the SD Card?
TMA has tested the CM-6010SD to see if this is possible.
We found that the data logger only logs one measurement at a time.
If you are measuring DC voltage, it logs to a file name dcvsxxx and if you are measuring DC current, it logs to a file called dcaxxx, and so on.
If you want to log both voltage and current at the same time, you will need to use a power meter such as the Lutron DW-6093.
However, this is AC only, so it will not suit every application.